|Hardware:||PC, OS Windows XP/Vista/7|
|Control stations:||Převodník USB/RS485 typ Dinel URC-485, Ultrazvukový hladinoměr Dinel ULM-53L s komunikací MODBUS|
|Date of commission:||2013|
The correct function of the level meter is signaled by a green light. There are other light signals warning the operator once the levels are outside measured range. There is a blue bargraph signaling the level position in the tank. Buttons on the side allow to open other panels.
The time progress of filling or draining is displayed in a graph on the right. Detailed view is opened after pushing the History button.
There are several buttons on the toolbar allowing to select the language, stop the application and set the MODBUS parameters.
The History panel allows to examine the details of filling/draining in time. It can also be used to export the data to Excel or csv and print the graph or table.
The basic setup panel is used in order to enter the range of connected ultrasound level meter and the corresponding volume of the tank.
The service setup is used for entering detailes measurement parameters. It is possible to enter the measuremenet units, number of measurements counted into average, temperature compensations, measurement period, etc. It is also possible to RESET the meter or return to the default settings.
These panels are used in order to enter the MODBUS communication parameters (SLAVE – ultrasound level meter) and it is also possible to read the information from the connected level meter (serial number, firmware version, sensor ID).